The product labeling information includes all published material associated to a drug. Product labeling documents include information like generic names, active ingredients, ingredient strength dosage, routes of administration, appearance, usage, warnings, inactive ingredients, etc.
Product Labeling Index
Instructions For Blood Collection Using Anticoagulant Citrate Phosphate Dextrose Solution, Usp (Cpd) Single And Multiple Blood-Pack® Units
- Rx onlyIntegral Donor Tube (IDT)Use aseptic technique.Precaution: Do not use unless anticoagulant is clear. 1.Identify Blood-Pack unit using appropriate donor identification system. Confirm that all numbered tubing of each Blood-Pack unit contains its own identical segment numbers. 2.Adjust donor scale to desired collection weight. 3.Position primary container on the donor scale as far as possible below donor arm and clamp donor tubing with hemostat. 4.Apply pressure to donor’s arm and disinfect site of venipuncture. 5.If blood pressure cuff is used, inflate to approximately 60 mmHg. 6.Remove needle cover per instructions below:(a)Holding the hub and cover near the tamper-evident seal, twist cover 1/4 turn to break seal.(b)Remove needle cover, being careful not to drag the cover across the needle point. Perform venipuncture, appropriately secure donor needle and/or tubing and release hemostat. 7.Mix blood and anticoagulant at several intervals during collection and immediately after collection. 8.Collect the appropriate volume based on Blood-Pack unit used. Note: The volume of anticoagulant is sufficient for the blood collection indicated on Blood-Pack unit ± 10%. 9.Apply hemostat to donor tubing. 10.Release pressure on the donor’s arm as appropriate and withdraw donor needle.Precaution: Use care to avoid accidental needle sticks when filling sample tubes. The Fenwal Needle/Tube Sampling Protector may be used to reduce the risk of needle stick injury. 11.Slide sample tube onto needle until stopper is punctured. Release hemostat to fill sample tube. 12.To fill additional tubes, apply a hemostat to donor tube and repeat step 11. 13.Strip blood from donor tubing into container, mix and allow the tubing to refill; repeat once. Seal at X marks on donor tubing to provide numbered aliquots of anticoagulated blood for typing or crossmatching. 14.Discard needle into an appropriate biohazardous waste container following established procedures. 15.Store suspended CPD Whole Blood/Red Blood Cells between 1 and 6° C. 16.Infuse CPD Whole Blood/Red Blood Cells within 21 days of collection.Dispose of container appropriately. Single use only.Store at Controlled Room Temperature. USP Definition of “Controlled Room Temperature” United States Pharmacopeia, General Notices United States Pharmacopeial Convention, Inc. 12601 Twinbrook Parkway, Rockville, MD.Fenwal, Inc.Lake Zurich, IL 60047 USA1-800-933-6925Made in USAFenwal International, Inc.Road 357, Km. 0.8Maricao, PR 00606– ManufacturerFENWAL and BLOOD-PACK are trademarks of Fenwal, Inc© 2008 Fenwal, Inc. All rights reserved.07-19-58-106 REV: B10/2008
Package/Label Display Panel
- Code 4R0837MC1 UnitFenwal™Anticoagulant Citrate Phosphate Dextrose Solution, USP (CPD) BLOOD-PACK™ UnitSingle For Collection and Processing of 250 mL BloodIntegral Donor Tube, 16 ga. Ultra Thin Wall NeedleRx onlyEach unit consists of a PL 146 Plastic primary container with 35 mL of CPD solution containing 921 mg Sodium Citrate (dihydrate) USP, 893 mg Dextrose (monohydrate) USP, 105 mg Citric Acid (anhydrous) USP and 78 mg Monobasic Sodium Phosphate (monohydrate) USP. pH may have been adjusted with sodium hydroxide.Sterile, non-pyrogenic fluid path.See instructions for use.Single use only.Dispose of container appropriately.FENWAL and BLOOD-PACK are trademarks of Fenwal, Inc.Store at Controlled Room Temperature (refer to direction insert). •Open pouch by tearing across at notch. •Direct handling of product surfaces prior to extended storage in the foil pouch, may result in mold growth. •Unused units in open foil pouch may be kept up to 60 days by folding and securing open end of foil pouch to prevent possible loss of moisture, provided:I)Units are not removed from foil pouch, orII)Unused units removed from foil pouch are returned to the foil pouch within 12 hours. Units may be removed from the pouch and returned only once. •Units removed from the foil pouch (that are not returned to the pouch within 12 hours) must be used within 4 days (96 hours). Units out of the foil pouch for longer than 96 hours must be discarded.Manufacturer:Fenwal, Inc.Lake Zurich, IL 60047 USAMade in USA07-28-05-458 REV: A
* Please review the disclaimer below.